Sales & Support
Request A Quote - Email
Payment & Shipping Terms:
DOA Benzodiazepines Rapid Test (BZO)
A rapid, one step test for the qualitative detection of Benzodiazepines in human urine.
For healthcare professionals including professionals at point of care sites.
For in vitro diagnostic use only.
MOR/MET/THC/AMP/MDMA/MTD/COC/TCA/BAR/BZO, up to 15 kinds of Drug Tests
Format: Strip, Cassette, Cup,and Panel
The BZO One Step Benzodiazepines Test Strip is a lateral flow chromatographic immunoassay for the detection of Oxazepam (major metabolite) in urine at a cut-off concentration of 300 ng/mL. This test will detect other Benzodiazepines, please refer to Analytical Specificity table in this package insert.
This assay provides only a preliminary analytical test result. A more specific alternate chemical method must be used in order to obtain a confirmed analytical result. Gas chromatography/mass spectrometry (GC/MS) is the preferred confirmatory method. Clinical consideration and professional judgment should be applied to any drug of abuse test result, particularly when preliminary positive results are used.
The BZO One Step Benzodiazepines Test Strip is an immunoassay based on the principle of competitive binding. Drugs which may be present in the urine specimen compete against the drug conjugate for binding sites on the antibody.
During testing, a urine specimen migrates upward by capillary action. Benzodiazepines, if present in the urine specimen below cut-off, will not saturate the binding sites of the antibodies in the test strip. The antibody coated particles will then be captured by immobilized benzodiazepine-protein conjugate and a visible colored line will show up in the test line region. The colored line will not form in the test line region if the benzodiazepines level exceeds cut-off concentration, because it will saturate all the binding sites of anti-benzodiazepines antibody.
POSITIVE: One colored line appears in the control region (C). No line appears in the test region (T). This positive result indicates that the Benzodiazepines concentration exceeds the detectable level.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test strip. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
Contact Person: Ms. Anna Lee