|Place of Origin:||China|
|Minimum Order Quantity:||100 Tests|
|Packaging Details:||40 tests/box|
|Delivery Time:||7 working days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Supply Ability:||1000000 pcs per month|
HBcAb Rapid Test Strip
Cat. No.: RH0210C
FOR THE QUALITATIVE ASSESSMENT OF HBcAb IN HUMAN SERUM, PLASMA OR WHOLE BLOOD
The One Step HBcAb Test is a rapid chromatographic immunoassay for the qualitative detection of Hepatitis B virus core antibody (HBcAb/anti-HBc) in Whole Blood /Serum / Plasma.
The One Step HBcAb Test is a competitive immunoassay. Colloidal gold conjugated anti-HBcAg antibody complexes are dry-immobilized in the test device. When the specimen is added, it migrates with the gold conjugate complexes by capillary diffusion through the strip. If present, HBcAb will compete with gold conjugate complexes for the limited amount HBcAg immobilized in the Test Zone (T). It will prevent the gold conjugate complexes from reacting with HBcAg and no red line appears in the Test Zone (T). If there is no HBcAb in the specimens, gold conjugate complexes will react with HBcAg and a visible red line appears. To serve as a procedural control, a red line will always appears in the Control Zone (C) which indicates the validity of the test.
Storage and Stability
Additional Special Equipment
Test devices Disposable specimen droppers
Package insert Buffer(for whole blood only)
Materials Required But Not Provided
Specimen collection containers
Specimen Collection and Preparation
Allow the test, specimen and/or controls to reach room temperature 15-30℃ (59-86℉) prior to testing.
Applying sufficient amount of specimen is essential for a valid test result. If migration (the wetting of membrane) is not observed in the test window after one minute, add one more drop of specimen to the specimen well.
Interpretation of Results
Positive: One colored line appears in the control region(C).No apparent colored line appear in the test line region.
Negative: Two lines appear. One line should always appear in the control line region(C), and another one apparent colored line should appear in the test line region.
Invalid: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test device. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
Span Biotech Ltd. is a research based company for rapid tests, with strong support from National Key Laboratory of Technology Projects of 10th and 11th Five-Year Plan and Faculty of Life Sciences of HuBei University. SpanBio also housed a R&D team that is developing gene recombination, cell cultivation and protein purification techniques. SpanBio pays strict attention on rapid tests for human being, animal diseases and food safety detection. It provides a number of customized services to professional distributors and partnering affiliates with excellent quality, competitive prices and super service.
Span Biotech Ltd.
Add: 6A, Unit 5, Building 3, Sunshine Aloha, Bantian Industrial Park, Banxuegang Road , Longgang Shenzhen 518129