|Place of Origin:||CHINA|
|Minimum Order Quantity:||K2|
|Delivery Time:||within 5 working days (depends on your quantity)|
|Payment Terms:||T/T,Western Union,Paypal|
One Step Synthetic Cannabis Test Device (Urine)
Format: Strip, Cassette, Cup,and Panel
The K2 One Step Synthetic Cannabis Test is an immunochromatography-based one step in vitro test. It is designed for qualitative determination of synthetic cannabis JWH-018 and JWH-073’s major metabolites in human urine specimens at cut-off level of 50 ng/ml. This assay has not been evaluated in the point of care location and is for forensic use only.
This assay provides only a preliminary analytical test result. A more specific alternative chemical method must be used in order to obtain a confirmed analytical result. Gas chromatography/mass spectrometry (GC/MS) has been established as the preferred confirmatory method by the Substance Abuse Mental Health Services Administration (SAMHSA). Clinical consideration and professional judgment should be applied to any drug of abuse test result, particularly when preliminary positive results are indicated.
The K2 One Step Synthetic Cannabis Test is based on the principle of specific immunochemical reaction between antibodies and antigens to analyze particular compounds in human urine specimen. The assay relies on the competition for binding antibody between drug conjugate and free drug which may be present in the urine specimen being tested. When drug is present in the urine specimen, it competes with drug conjugate for the limited amount of antibody-dye conjugate. When the amount of drug is equal or more than the cut-off it will prevent the binding of drug conjugate to the antibody. Therefore, a positive urine specimen will not show a colored band on the test line zone, indicating a positive result, while the presence of a colored band indicates a negative result.
A control line is present in the test window to work as procedural control. This colored band should always appear on the control line zone if the test device is stored in good condition and the test is performed appropriately.
POSITIVE: Only one colored band appears in the control region (C). No apparent colored band appears in the test region (T).
NEGATIVE RESULT:Two colored bands appear on the membrane. One band appears in the control region (C) and another band appears in the test region (T).
INVALID RESULT:Control band fails to appear. Results from any test which has not produced a control band at the specified reading time must be disgarded. Please review the procedure and repeat with a new test. If the problem persists, discontinue using the kit immediately and contact your local distributor.
1.The K2 One Step Synthetic Cannabis Test Device (Urine) provides only a qualitative, preliminary analytical result. A secondary analytical method must be used to obtain a confirmed result. Gas chromatography/mass spectrometry (GC/MS) is the preferred confirmatory method. 1,2
2. It is possible that technical or procedural errors, as well as other interfering substances in the urine specimen may cause erroneous results
3. Adulterants, such as bleach and/or alum, in urine specimens may produce erroneous results regardless of the analytical method used. If adulteration is suspected, the test should be repeated with another urine specimen.
4. A positive result indicates presence of the drug or its metabolites but does not indicate level of intoxication, administration route or concentration in urine.
5. A negative result may not necessarily indicate drug-free urine. Negative results can be obtained when drug is present but below the cut-off level of the test.
6. Test does not distinguish between drugs of abuse and certain medications.