|Place of Origin:||CHINA|
|Minimum Order Quantity:||50 TESTS|
|Packaging Details:||40 tests/box|
|Delivery Time:||7 working days|
|Payment Terms:||T/T, Western Union, MoneyGram|
Canine Distemper Virus Ag Test is a sandwich lateral flow immunochromatographic assay for the qualitative detection of canine Distemper virus (CDV Ag) in dog’s secretions or serum.
Assay Time: 5-10 min
Sample: secretions or serum
Canine Distemper Virus Ag Test is based on sandwich lateral flow immunochromatographic assay. The test device has a testing window. The testing window has an invisible T (test) zone and C (control) zone. When sample is applied into the sample hole on the device, the liquid will laterally flow on the surface of the test strip. If there is enough Distemper antigen in the sample, a visible T band will appear. The C band should always appear after a sample is applied, indicating a valid result. By this means, the device can accurately indicate the presence of Canine Distemper antigen in the sample.
1. Foil pouches, each containing one cassette, one pipette and a desiccant.
2. Assay buffer tubes (0.7 mL each)
3. Swab sticks
4. Product Manual
1. Collect dog’s ocular and nasopharyngeal secretions with the swab stick. Make the swab wet sufficiently.
2. Insert the wet swab into the provided assay buffer tube. Agitate it to assure good sample extraction.
3. If using serum sample, please do a dilution of 1:2 with the provided assay buffer.
4. Take out the cassette from the foil pouch and place it horizontally.
5. Gradually drip 3 drops of sample extraction into the sample hole.
6. Interpret the result in 5-10 minutes. Result after 10 minutes is considered as invalid.
INTERPRETATION OF RESULTS
Positive: The presence of both C band and T band, no matter T band is clear or vague.
Negative: Only clear C band appears.
Invalid: No colored band appears in C zone, no matter whether T band appears.
The kit can be stored at room temperature (2-30°C). The test kit is stable through the expiration date (18 months) marked on the foil pouch. DO NOT FREEZE. Do not store the test kit in direct sunlight.
1. For best results, please strictly adhere to these instructions.
2. All reagents must be at room temperature before running the assay.
3. Do not remove test cassette from its pouch until immediately before use.
4. Do not reuse the test kit.
5. Do not use the test beyond its expiration date marked on the foil pouch.
6. The components in this kit have been quality control tested as standard batch unit. Do not mix components from different lot numbers.
Canine Distemper Virus Ag Test is for in vitro veterinary diagnosis use only. All results should be considered with other clinical information available from veterinarian. For an accurate result, It is suggested to apply other method such as PCR for final determination in practice.